Portosystemic shunts are common hepatic vascular anomalies in small animal patients.1–3 Medical management of PSSs has been associated with chronic hepatic fibrosis and portal hypertension, which can result in progressive hepatic failure, as well as clinical signs of persistent or progressive neurologic and urinary tract abnormalities.4–7 The long-term prognosis is generally more favorable following treatment with gradual surgical attenuation of a PSS, compared with that following medical management, provided the patient survives the immediate postoperative period.6–10
Various methods have been used for PSS attenuation including silk ligatures,11 intravascular thrombogenic coils,11,12 hydraulic occluders,4,13 cellophane banding,8,11,14 and ameroid constrictors.2,15,16 Among those devices, ameroid constrictors have gained popularity because they gradually occlude the PSS, provide predictable closure, and are associated with low surgical and postoperative complications.1,15,16 Ameroid is a hygroscopic, compressed casein that undergoes expansion when immersed in fluid, which is characterized by rapid expansion initially followed by a period of slower expansion.1,15,17
Investigators of a retrospective study14 that involved 106 dogs and 5 cats with congenital PSSs in which the shunt was surgically attenuated by use of a cellophane band suggest that the initial constriction of the PSS should not be < 2.0 to 3.0 mm. Because scientifically sound recommendations regarding the appropriate size of ameroid constrictor for use in attenuation of PSSs with particular diameters are lacking, the decision of which size of constrictor to use for attenuation of each PSS is subjective. Surgeons may handle ameroid constrictors of multiple sizes before choosing which size of constrictor to use during a PSS attenuation procedure. Consequently, for the purpose of frugality and because the manufacturer does not advise against multiple sterilizations, unused ameroid constrictors are frequently resterilized after surgical procedures. The casein material in ameroid constrictors is a type of protein. In 1 study,18 the use of heat to sterilize micellar casein concentrates caused proteolysis and protein dissociation. Thus, resterilization of ameroid constrictors could cause denaturation of the casein material and increase its permeability, which could affect the rate of closure of the constrictors when used for PSS attenuation.
The objective of the study reported here was to evaluate the lumen area of ameroid constrictors after ≥ 1 HPGP sterilization procedure. Our hypothesis was that the number of times that an ameroid constrictor undergoes HPGP sterilization would have no effect on its final lumen area.
Hydrogen peroxide gas plasma
Ameroid constrictor, Research Instruments NW, Lebanon, Ore.
VHP MD140X biodecontamination unit, STERIS Corp, Mentor, Ohio.
VITROS 5,1 FS, Ortho Clinical Diagnostics Inc, Rochester, NY.
Chocolate agar plate, Remel Inc, Lenexa, Kan.
Class II Type A/B3 biological safety cabinet, NuAire, Plymouth, Minn.
IR Autoflow CO2 water jacketed incubator, NuAire, Plymouth, Minn.
Microsoft Excel, version 14.06129.5000, Microsoft Corp, Redmond, Wash.
Olympus Tough TG-820 12-megapixel digital camera, Olympus Corporation of the Americas, Tokyo, Japan.
NIH Image J, National Institutes of Health, Bethesda, Md. Available at: rsb.info.nih.gov/nih-image. Accessed Nov 1, 2013.
STATA SE, version 12.1, StataCorp, College Station, Tex.
1. Kyles AE, Hardie EM, Mehl M, et al. Evaluation of ameroid ring constrictors for the management of single extrahepatic portosystemic shunts in cats: 23 cases (1996–2001). J Am Vet Med Assoc 2002;220:1341–1347.
3. Tobias KM, Rohrbach BW. Association of breed with the diagnosis of congenital portosystemic shunts in dogs: 2,400 cases (1980–2002). J Am Vet Med Assoc 2003;223:1636–1639.
4. Adin CA, Sereda CW, Thompson MS, et al. Outcome associated with use of a percutaneously controlled hydraulic occluder for treatment of dogs with intrahepatic portosystemic shunts. J Am Vet Med Assoc 2006;229:1749–1755.
5. Johnson CA, Armstrong PJ, Hauptman JG. Congenital portosystemic shunts in dogs: 46 cases (1979–1986). J Am Vet Med Assoc 1987;191:1478–1483.
6. Hottinger HA, Walshaw R, Hauptman JG. Long-term results of complete and partial ligation of congenital portosystemic shunts in dogs. Vet Surg 1995;24:331–336.
7. Greenhalgh SN, Dunning MD, McKinley TJ, et al. Comparison of survival after surgical or medical treatment in dogs with a congenital portosystemic shunt. J Am Vet Med Assoc 2010;236:1215–1220.
8. Frankel D, Seim H, MacPhail C, et al. Evaluation of cellophane banding with and without intraoperative attenuation for treatment of congenital extrahepatic portosystemic shunts in dogs. J Am Vet Med Assoc 2006;228:1355–1360.
10. Bright SR, Williams JM, Niles JD. Outcomes of intrahepatic portosystemic shunts occluded with ameroid constrictors in nine dogs and one cat. Vet Surg 2006;35:300–309.
11. Youmans KR, Hunt GB. Experimental evaluation of four methods of progressive venous attenuation in dogs. Vet Surg 1999;28:38–47.
12. Léveillé R, Johnson SE, Birchard SJ. Transvenous coil embolization of portosystemic shunt in dogs. Vet Radiol Ultrasound 2003;44:32–36.
13. Sereda CW, Adin CA. Methods of gradual vascular occlusion and their applications in treatment of congenital portosystemic shunts in dogs: a review. Vet Surg 2005;34:83–91.
14. Hunt GB, Kummeling A, Tisdall PLC, et al. Outcomes of cellophane banding for congenital portosystemic shunts in 106 dogs and 5 cats. Vet Surg 2004;33:25–31.
15. Vogt JC, Krahwinkel DJ, Bright RM, et al. Gradual occlusion of extrahepatic portosystemic shunts in dogs and cats using the ameroid constrictor. Vet Surg 1996;25:495–502.
16. Mehl ML, Kyles AE, Hardie EM, et al. Evaluation of ameroid ring constrictors for the treatment for single extrahepatic portosystemic shunts in dogs: 168 cases (1995–2001). J Am Vet Med Assoc 2005;226:2020–2030.
17. Adin CA, Gregory CR, Kyles AE, et al. Effect of petrolatum coating on the rate of occlusion of ameroid constrictors in the peritoneal cavity. Vet Surg 2004;33:11–16.
18. Beliciu CM, Sauer A, Moraru CI. The effect of commercial sterilization regimens on micellar casein concentrates. J Dairy Sci 2012;95:5510–5526.
19. Lange PE, Seivers HH, Nurnberg JH, et al. A new device for slow progressive narrowing of vessels. Basic Res Cardiol 1985;80:430–435.
20. Besancon MF, Kyles AE, Griffey SM, et al. Evaluation of the characteristics of venous occlusion after placement of an ameroid ring constrictor in dogs. Vet Surg 2004;33:597–605.
22. Monnet E, Rosenberg A. Effect of protein concentration on rate of closure of ameroid constrictors in vitro. Am J Vet Res 2005;66:1337–1340.
23. Murphy ST, Ellison GW, Long M, et al. A comparison of the ameroid constrictor versus ligation in the surgical management of single extrahepatic portosystemic shunts. J Am Anim Hosp Assoc 2001;37:390–396.